Using the Capacitor Wizard for the First Time
This article is in response to new customer feedback. It was brought to my attention
that the Capacitor Wizard operating manual needed a "getting started" section
for first time users. I decided to write a whole article on the subject. Measuring for ESR
(Equivalent Series Resistance) may be new to many techs so lets briefly review ESR and its
detrimental effects on various electronic circuits.
ESR is the DYNAMIC pure resistance of a capacitor to an AC signal. High ESR can cause time
constant problems, capacitor heating, circuit loading, total circuit failure etc. A
switching power supply may not reliably start - or start at all. Slight hum bars appear in
the video of a VCR or monitor. A TV may be pulled in from the sides/top/bottom. Partial
loss of a color, intermittent operation, diode and transistor failure over a period of
time. These problems and many more are often caused by capacitors with normal capacitance
but high ESR.
Why do I need the Capacitor Wizard? - why not use a standard capacitance checker or a DC
ohm meter? ESR does not exist as a STATIC quantity therefore it cannot be measured by a
conventional capacitance meter or a DC ohm meter. ESR exists only when alternating current
is applied to a capacitor or when a capacitor's dielectric charge is changing states. ESR
can be considered to be the TOTAL IN PHASE AC resistance of a capacitor. ESR includes the
DC resistance of the leads, DC resistance of the connections to the dielectric, capacitor
plate resistance, and the IN PHASE AC resistance of the dielectric material at a
particular frequency and temperature. ESR is affected by every capacitor physical quantity
- so any problem with a capacitor (except a short) shows up as an increase in ESR! The
combination of components that make up ESR are symbolized by a resistor in series with a
capacitor as shown above. This symbolic resistor does not really exist as a physical
entity so direct measurements across the ESR resistor are not possible! However, if a
method of correcting for the effects of capacitive reactance is provided, and considering
all ESR resistances are IN PHASE, ESR can be calculated and measured by using the basic
electronics formula E = I x R! This is the Basic Electronics foundation used to design the
From the above definition of ESR you should now understand that any capacitor failure mode
will cause the ESR of that capacitor to increase in resistance (except shorts). If a
capacitor has dried out, is open, or otherwise has suffered DRASTIC failure, the measured
ESR wig be near inanity. If a capacitor has changed capacitance slightly it has a problem
and that problem will show up as a several ohms increase in the ESR. For this reason a
standard capacitance meter that does not measure ESR could lead you t o conclude that a
bad capacitor is actually good! ! I have had many reports from Capacitor Wizard owners
that have found bad caps using the Capacitor Wizard that other cap testers found good -
and proved it by replacing the cap!
Get experience measuring known bad caps: OK, lets test some caps. As a new user of the
Capacitor Wizard I highly recommend that you test some known bad caps to see how they
respond and build your confidence in the instrument. The Capacitor Wizard probes are nonpolar so don't worry about polarity. You will probably have mostly open capacitors and
they will barely move the meter. Get a 100 ohm resistor and put it in parallel with a bad
open cap. Notice that the meter still just barely moves! Try different values of
resistance and see how the meter responds. Get a transistor or a diode and try to measure
across the junctions. You will notice the Capacitor Wizard does NOT respond. That is
because the Capacitor Wizard test signal is only about 5 millivolts RMS. It takes 300 to
600 millivolts to turn on solid state devices. Get a transformer or a choke similar to
what you expect to find in the equipment you repair. Measure across the primary, secondary
or anywhere you like. You will notice that the Capacitor Wizard does not respond. That is
because the INDUCTIVE REACTANCE at the 100 kHz test frequency of the Capacitor Wizard is
many orders of magnitude larger than the resistance range of the Capacitor Wizard. A high
percentage of your bad caps will fall into this category - OPEN. This demonstration proves
that open caps are very easy to find IN CIRCUIT and the associated circuitry has little
effect on the Capacitor Wizard operation IN CIRCUIT.
You probably noticed that not all of your bad caps checked open. I'm sure that some of the
caps probably rang the "Cap Good" beeper (ESR less than 1/2 ohm). Those caps are
more than likely GOOD. Every shop accidentally replaces a good cap now and then. Just to
make sure they aren't shorted, check them with a standard ohm meter. Check them on your
standard out of circuit capacitance reading meter. You'll probably find that they are all
good. Shorted caps are pretty rare. I've got boxes of bad caps from repair shops around
town and have yet to find a shorted cap.
Well, finding those open caps and good caps was easy. Now lets look at other bad caps that
require a little more experience with the Capacitor Wizard and some knowledge about
capacitor TYPES and USES. You probably found caps from 1 to 30 ohms ESR in your bad box.
How do you tell the good caps from the bad??
Whether the ESR of a particular capacitor is correct or too high can always be determined
by comparing the suspicious capacitor to a known good one of the same value, voltage
rating, and type. Unfortunately one doesn't always have another capacitor to compare
against. Experience is the best teacher here, however there are some general guidelines:
The higher the rated working voltage, the higher the normal ESR
Capacitors used in power Switching applications need to have really LOW ESR - less than
Nonpolar Caps are normally less than 1/2 ohm
The next logical question about ESR is "How HIGH is TOO HIGH"? That's a judgment
call that can only be based on experience or comparison to a known good cap (or access to
the engineering data from the capacitor or equipment manufacturer - ha ha!). Over 10 ohms
is certainly too high for most applications. Over 3 ohms is too high for Horiz/Vert
switching applications. Over 1/2 ohm is too high for power switching applications. By
comparison you will gain experience and know when to be suspicious. These are my opinions.
Here are some actual repair situations:
Example: 47uf @50vdc measures 25 ohms ESR in circuit - BAD CAP The suspect capacitor is a
47uf 850vdc in a switching power supply for a VCR. The Capacitor Wizard has measured 25
ohms ESR in circuit. That is higher than 15 ohms and much to high for any quality cap. A
new capacitor measured 5 ohms ESR. The new capacitor fixed the VCR. In my opinion the new
capacitor was not of the highest quality (5 ohms is too high) however it did fix the VCR.
The use of these low quality inexpensive import capacitors is probably the reason we see
so much capacitor failure in consumer electronic equipment! A higher quality cap with a
lower ESR of the same kind costs more money but will measure less than 1 ohm and be more
reliable. Conclusion: This is a higher voltage capacitor and can be expected to normally
measure higher than 1/2 ohm. In my judgment any "switch mode" capacitor that
measures more than 3 ohms ESR is suspect no matter what the voltage rating. However you
may obviously get by with the 5 ohms ESR in that particular circuit. For comparison, the
bad part was checked "out of circuit" on a well known competitors S2000 Cap
analyser and it determined that the cap was GOOD even though the ESR measured 25 ohms!
That manufacturer made a huge mistake by trying to calculate good and bad ESR from entered
and measured data. It can't be done reliably. That is why we don't simply have a good/bad
indication on our meter scale. Any cap over 3 ohms is suspect. This is my Experience.
Example: 1000uf @ 6vdc measures 1.5 ohms in circuit - BAD CAP This is a
little brown 1000uf 6vdc cap used in lots of VCR switching power supplies. The Capacitor
Wizard measured 1.5 ohms in circuit. Because the capacitors operating voltage is so low
(6vdc) and its used in a switching power supply, I would expect a normal ESR reading of
less than 1/2 ohm. Comparison with a known good cap confirmed it should measure less than
1/2 ohm. Replacing this cap cured the trouble. This particular cap goes bad often as I
have many in my box of bad caps gathered from local repair companies. If you work on VCRs,
I bet you have some too.
Summary: Measuring ESR is a very good indicator of capacitor failure. For switch mode
circuits it is the ONLY reliable capacitor test, IN or OUT of circuit!. Open caps and caps
with really high ESR (over 10 ohms) are easy to find in circuit and need to be replaced.
Marginal caps that measure between 1 and 10 ohms ESR require some experience with the
Capacitor Wizard and/or comparison to a known good cap of the same voltage, value, and
type. Caps above an operating voltage of 35vdc have a normally higher ESR (around 1 to 3
ohms) than caps of a lower voltage (less than 1/2 ohm ESR).
I know of no perfect formula or rule that can always tell normal ESR from marginal ESR
other than comparison with a known good part. The obvious solution is to obtain the
capacitor manufacturers data manuals on the EXACT capacitor measured but that is not
normally practical. As a technician myself I always follow this rule: "If in doubt,
replace". You will eliminate a lot of recalls and cure many weird and undefinable
intermittent problems if you follow this rule.
President, Midwest Devices, Inc.
Using The Capacitor Wizard - First Time